In July 2019, the U.S. Food and Drug Administration (FDA) issued alerts on Medtronic’s 600-series MiniMed insulin pumps.
It appears that when air pressure changes quickly, such as during take-off or landing, the device’s keypad buttons could temporarily become “unresponsive”.
This means that users may not be able to program a bolus or suspend delivery, since the buttons won’t temporarily press down. However, the basal insulin continues to be delivered.
The good news is that in most cases, when atmospheric pressures normalizes, the buttons will work again in about thirty minutes. Also, pump users can resolve the issue by taking off the battery cap and putting it back on again.
Medtronic is actively working on a long-term hardware solution for this rare situation.
The alerts, classified as a class II recall by the FDA, apply to the MiniMed 620G, 630G, 640G and 670G models. A class II recall is considered medium severity and is used when the probability of a serious adverse consequence is considered remote. The FDA and Medtronic are not requesting any devices be returned, they are just making sure Medtronic pump wearers are aware of this issue.
Read the complete article here.
New Technology Toolkit – Earn 3.0 CEs Premiers August 20 & 23
When it comes to insulin pumps, sensors and calculation, many of us feel overwhelmed and unsure about diabetes technology management. Plus, with the vast amount of information, it may seem impossible to figure out what to focus on for our clinical practice and to prepare for the diabetes certification exam.
If you want cutting edge information on diabetes technology, problem solving and using formulas to determine appropriate insulin dosing, we highly recommend this toolkit.
Sign up for Diabetes Blog Bytes – we post one daily Blog Byte from Monday to Friday. And of course, Tuesday is our Question of the Week. It’s Informative and FREE! Sign up below!
October 28 @ 7:30 am – 4:00 pm
April 8, 2020 – April 10, 2020
November 12 @ 11:30 am – November 14 @ 12:45 pm