Medtronic released an update last week for its MiniMed 600 series pumps that we wanted to share with you.
You may remember our recent Blog Byte covering Medtronic’s MiniMed 600 Series Class I recall due to the two model’s pumps’ reservoir retainer rings being broken or missing. These issues resulted in an over or under-delivery of insulin. The pump models are:
Minimed’s updated info also includes helpful photos and detailed information to identify potential issues with the pumps and who to contact if issues are found. They also stress the importance of careful inspection of pump retainer ring on a regular basis.
Customers who have found an issue with their pump or who need additional information or support should call the 24-hour Medtronic Technical Support at 877-585-0166 or visit their website by clicking here.
Sign up for Diabetes Blog Bytes – we post one daily Blog Byte from Monday to Friday. And of course, Tuesday is our Question of the Week. It’s Informative and FREE! Sign up below!
May 13 – May 15
April 23 @ 8:30 am – April 30 @ 10:00 am
May 20 @ 11:30 am – 12:40 pm